|Hold handle and lower sensor into water without overflowing into the coupler|
BackgroundI needed turbidity values for a project I was working on so I decided to make an instrument that I could use universally.
The meter consists of:
|The sensor waterproof housing and base|
|Sensor mounted to base which is glued to coupler|
|Electronics acrylic covering and wiring|
The photos shows the approach which was to house the sensor down in a waterproof cylinder that is lowered into the water. The acrylic handle houses the electronics.
- A PVC coupler is the housing.
- A circular acrylic base was cut and is glued to the bottom of the coupler with "Plumbers Goop (PG)" (my favorite glue :)).
- The base has hole drilled that fits the sensor and the sensor has an Oring fitted and then is screwed with sheet metal screws to the base. The screws are sealed with PG as well.
- The base is glued to a 90 degree acrylic bracket with acrylic cement.
- The housing is glued to the bracket with PG.
- The bracket is fitted with an acrylic cover to provide a handle and some protection for the electronics.
- The acrylic peices were bent on my "acrylic bender".
The analog output of the turbidity sensor board is connected to the Feathers A0 analog input pin.
Power to the turbidity sensor board is provided by the VBUS power on the Feather.
I used the code that Adafruit provided on the tutorial site using the blueart_cmdmode sketch.
Phone appI used the Adafruit Bluefruit LE Connect mobile application.
Assy and testI simply followed the Adafruit tutorial for reading analog values using the mobile app and had no problems getting a continuous reading to my phone from the turbidity sensor.
Make sure that you have the switch on the sensor interface board in the analog position.